<!DOCTYPE html PUBLIC "-//W3C//DTD XHTML 1.0 Transitional//EN" "http://www.w3.org/TR/xhtml1/DTD/xhtml1-transitional.dtd">
<html xmlns="http://www.w3.org/1999/xhtml" xml:lang="pt-br" lang="pt-br"><head>
  <!-- base href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html" -->
  <meta http-equiv="content-type" content="text/html; charset=UTF-8">
  <meta name="robots" content="index, follow">
  <meta name="keywords" content="multiteste, multimetro, analógico, digital, resistência, corrente elétrica, tensão, teste, pilha, bateria, corrente alternada, corrente continua">
  <meta name="title" content="Multímetro analógico">
  <meta name="author" content="Olair Marcos">
  <meta name="description" content="Aprenda a utilizar o multímetro analólico com segurança em medições de resistência, tensão AC, tensão DC, corrente elétrica. Como ajustar e interpretar a leitura na escala.">
  <meta name="generator" content="Joomla! 1.5 - Open Source Content Management">
  <title>Multímetro analógico</title>
  <link href="http://www.eletronicadigital.com/site/component/jcomments/?task=rss&amp;object_id=6&amp;object_group=com_content&amp;tmpl=component" rel="alternate" type="application/rss+xml" title="Multímetro analógico">
  <link href="http://www.eletronicadigital.com/site/templates/rhuk_milkyway/favicon.ico" rel="shortcut icon" type="image/x-icon">
  <link rel="stylesheet" href="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/style.css" type="text/css">
  <link rel="stylesheet" href="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/fbox.css" type="text/css">
  <script src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/ca-pub-0131178064072026.js" type="text/javascript" async=""></script><script type="text/javascript" src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/jcomments-v2.js"></script>
  <script type="text/javascript" src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/ajax.js"></script>
  <script type="text/javascript" src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/mootools.js"></script>
  <script type="text/javascript" src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/caption.js"></script>
  <script type="text/javascript" src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/fbox_conf.js"></script>
  <script type="text/javascript" src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/fbox_engine-min.js"></script>


<link rel="stylesheet" href="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/system.css" type="text/css">
<link rel="stylesheet" href="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/general.css" type="text/css">
<link rel="stylesheet" href="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/template.css" type="text/css">
<link rel="stylesheet" href="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/orange.css" type="text/css">
<link rel="stylesheet" href="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/black_bg.css" type="text/css">
<!--[if lte IE 6]>
<link href="/site/templates/rhuk_milkyway/css/ieonly.css" rel="stylesheet" type="text/css" />
<![endif]-->
<meta name="google-site-verification" content="qxGohDOMqcqI5ZCQQUYOl-whznef7lV1B86b4DqR_rU">
</head>
<body style="" id="page_bg" class="color_orange bg_black width_fmax">
<a name="up" id="up"></a>
<div class="center" align="center">
	<div id="wrapper">
		<div id="wrapper_r">
			<div id="header">
				<div id="header_l">
					<div id="header_r">
						<div id="logo">
						<a href="http://www.eletronicadigital.com/site/" class="nounder"><img src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/blank.png" alt="" id="logo" border="0"></a>
						</div>
						
					</div>
				</div>
			</div>

			<div id="tabarea">
				<div id="tabarea_l">
					<div id="tabarea_r">
						<div id="tabmenu">
						<table class="pill" cellpadding="0" cellspacing="0">
							<tbody><tr>
								<td class="pill_l">&nbsp;</td>
								<td class="pill_m">
								<div id="pillmenu">
									
								</div>
								</td>
								<td class="pill_r">&nbsp;</td>
							</tr>
							</tbody></table>
						</div>
					</div>
				</div>
			</div>

			<div id="search">
				<form action="index.php" method="post">
	<div class="searchsearch">
		<input name="searchword" id="mod_search_searchword" maxlength="30" alt="Pesquisa" class="inputboxsearch" size="30" value="pesquisar..." onblur="if(this.value=='') this.value='pesquisar...';" onfocus="if(this.value=='pesquisar...') this.value='';" type="text">	</div>
	<input name="task" value="search" type="hidden">
	<input name="option" value="com_search" type="hidden">
	<input name="Itemid" value="9" type="hidden">
</form>
			</div>

			<div id="pathway">
				<span class="breadcrumbs pathway">
<a href="http://www.eletronicadigital.com/site/" class="pathway">Início</a> <img src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/arrow.png" alt=""> <a href="http://www.eletronicadigital.com/site/instrumentos-medicao.html" class="pathway">Instrumentos de medição</a> <img src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/arrow.png" alt=""> Multímetro analógico</span>

			</div>

			<div class="clr"></div>

			<div id="whitebox">
				<div id="whitebox_t">
					<div id="whitebox_tl">
						<div id="whitebox_tr"></div>
					</div>
				</div>

				<div id="whitebox_m">
					<div id="area">
									

						<div id="leftcolumn">
															<div class="module">
			<div>
				<div>
					<div>
											
<!-- /mod_html version 1.0.0.Alpha1-J1.5 (c) www.fijiwebdesign.com -->
 <center>
<script type="text/javascript"><!--
google_ad_client = "pub-0131178064072026";
//aranha-ceu160x600, menu-right, all, 27/01/08, eletdigital
google_ad_slot = "0783979667";
google_ad_width = 160;
google_ad_height = 600;
//--></script>
<script type="text/javascript" src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/show_ads.js">
</script><ins id="aswift_0_expand" style="display:inline-table;border:none;height:600px;margin:0;padding:0;position:relative;visibility:visible;width:160px;background-color:transparent"><ins id="aswift_0_anchor" style="display:block;border:none;height:600px;margin:0;padding:0;position:relative;visibility:visible;width:160px;background-color:transparent"><iframe marginwidth="0" marginheight="0" vspace="0" hspace="0" allowtransparency="true" scrolling="no" allowfullscreen="true" onload="var i=this.id,s=window.google_iframe_oncopy,H=s&amp;&amp;s.handlers,h=H&amp;&amp;H[i],w=this.contentWindow,d;try{d=w.document}catch(e){}if(h&amp;&amp;d&amp;&amp;(!d.body||!d.body.firstChild)){if(h.call){setTimeout(h,0)}else if(h.match){try{h=s.upd(h,i)}catch(e){}w.location.replace(h)}}" id="aswift_0" name="aswift_0" style="left:0;position:absolute;top:0;" frameborder="0" height="600" width="160"></iframe></ins></ins>
</center>

<!-- mod_html version 1.0.0.Alpha1-J1.5/ -->
					</div>
				</div>
			</div>
		</div>
			<div class="module">
			<div>
				<div>
					<div>
											
<!-- /mod_html version 1.0.0.Alpha1-J1.5 (c) www.fijiwebdesign.com -->
 <center>
<a href="http://www.mega-busca.com/lojas" target="_blank"><img src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/selomega-busca.png" alt="mega-BUSCA Brasil"></a>
</center>

<!-- mod_html version 1.0.0.Alpha1-J1.5/ -->
					</div>
				</div>
			</div>
		</div>
	
												</div>

												<div id="maincolumn">
													
							<table class="nopad">
								<tbody><tr valign="top">
									<td>
										<table class="contentpaneopen">
<tbody><tr>
		<td class="contentheading" width="100%">
				<a href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html" class="contentpagetitle">
			Multímetro analógico</a>
			</td>
				
		
					</tr>
</tbody></table>

<form method="post" action="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html"><span class="content_rating">Avaliação do Usuário:<img src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/rating_star.png" alt=""><img src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/rating_star.png" alt=""><img src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/rating_star.png" alt=""><img src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/rating_star.png" alt=""><img src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/rating_star.png" alt="">&nbsp;/&nbsp;658</span>
<br>
<span class="content_vote">Pior<input alt="vote 1 star" name="user_rating" value="1" type="radio"><input alt="vote 2 star" name="user_rating" value="2" type="radio"><input alt="vote 3 star" name="user_rating" value="3" type="radio"><input alt="vote 4 star" name="user_rating" value="4" type="radio"><input alt="vote 5 star" name="user_rating" value="5" checked="checked" type="radio">Melhor&nbsp;<input class="button" name="submit_vote" value="Avaliar" type="submit"><input name="task" value="vote" type="hidden"><input name="option" value="com_content" type="hidden"><input name="cid" value="6" type="hidden"><input name="url" value="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html" type="hidden"></span></form><table class="contentpaneopen">

<tbody><tr>
	<td class="createdate" valign="top">
		Qua, 31 de Março de 2010 17:30	</td>
</tr>


<tr>
<td valign="top">
	
	<table class="contenttoc" cellpadding="0" cellspacing="0">
	<tbody><tr>
		<th>Índice do Artigo</th>
	</tr>
	
	<tr>
		<td>
		<a href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html" class="toclink active">Multímetro analógico</a>
		</td>
	</tr>
	
			<tr>
				<td>
				<a href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html?start=1" class="toclink">Antes de começar</a>
				</td>
			</tr>
			
			<tr>
				<td>
				<a href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html?start=2" class="toclink">Utilizando o multímetro</a>
				</td>
			</tr>
			
		<tr>
			<td>
				<a href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html?showall=1" class="toclink">Todas as Páginas</a>
			</td>
		</tr>
		</tbody></table><div class="pagenavcounter">Página 1 de 3</div><a href="http://www.eletronicadigital.com/site/user/images/multimetro-analogico/multimetro-analogico-sanwa-475px.jpg" target="_blank">
<span class="frontbox" name="fbox_6"></span>
<img style="margin: 5px; float: left;" title="Multímetro analógico Sanwa" alt="multimetro-analogico-sanwa-475px" src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/multimetro-analogico-sanwa-475px_118x173_875f7bcb6fa4c0a87e2.jpg" height="173" border="0" width="118">
</a>O multímetro analógico também conhecido como multiteste analógico, é
 um instrumento indispensável para quem deseja realizar experimentos no 
segmento da eletroeletrônica.<br> Com este instrumento de medição 
podemos comprovar as três grandezas da lei de OHM, como medir a tensão 
elétrica, a corrente elétrica, a resistência elétrica e ainda testar 
diversos componentes eletrônicos como resistores, capacitores, diodos, 
transformadores, leds, transistores e outros.<br> <br> Conheça as principais funções do multímetro analógico neste artigo.<br> <br> 



<br> Em lojas de instrumentos de medição vamos encontrar uma grande 
variedade de multímetros analógicos e digitais, portáteis ou de bancada.<br>
 Neste artigo vamos descrever sobre os multímetros analógicos utilizados
 em oficinas de consertos e cursos de eletrônica. Os preços podem variar
 de acordo com a marca e a quantidade de recursos que cada equipamento 
tiver, alguns mais comuns com poucas escalas, outros mais completos, com
 mais escalas, mais sensíveis e precisos.<br> Todos trazem as funções básicas de medição de tensão, corrente e resistência elétrica.<br>
<div style="text-align: center"><img src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/multimetro-analogico.jpg" alt="multimetro-analogico" title="Multímetros analógicos" style="margin: 5px; width: 475px; height: 250px;" height="250" border="0" width="475"></div>
<br> <span style="font-size: medium"><strong>Como funciona o multímetro analógico</strong></span><span style="text-decoration: underline;"><br> <br> </span>Trata-se
 de um instrumento que possui um ponteiro montado sobre uma bobina 
móvel, a bobina móvel está fixada no meio de um campo magnético 
constituído de um imã permanente.<br> No momento que uma corrente 
elétrica percore o enrolamento da bobina móvel surge um campo magnético 
na bobina, que interage com o campo magnético do ímã, dependendo do 
sentido da corrente elétrica o ponteiro poderá se movimentar para 
direita ou para esquerda na escala do instrumento.<br> Quando o 
instrumento está sem uso, o ponteiro estará em seu ponto de descanso, 
totalmente a esquerda da escala, ao realizar uma medição o ponteiro 
deverá se movimentar para a direita na escala. Se a movimentação do 
ponteiro for para a esquerda entendemos que a polaridade das pontas em 
relação ao ponto de medição está invertido.<br> Assim podemos afirmar 
que o nosso multímetro analógico é polarizado, então devemos tomar o 
cuidado para sempre utilizar a ponta vermelha no (+) positivo e a ponta 
preta (-) no negativo ao ponto de medição.<br> <br> 
<br><div class="pagenavbar"><div>Página anterior - <a href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html?start=1">Página seguinte &gt;&gt;</a></div></div><br></td>
</tr>

<tr>
	<td class="modifydate">
		Última atualização em Ter, 25 de Maio de 2010 19:59	</td>
</tr>
</tbody></table>
<span class="article_separator">&nbsp;</span>
<script type="text/javascript">
var jcomments=new JComments(6, 'com_content','http://eletronicadigital.com/site/index.php?option=com_jcomments&amp;tmpl=component');
jcomments.setList('comments-list');
</script>
<div id="jc">
<div id="comments"><h4>Comentários<a class="rss" href="http://www.eletronicadigital.com/site/component/jcomments/?task=rss&amp;object_id=6&amp;object_group=com_content&amp;tmpl=component" title="Sina RSS para comentários ao tópico." target="_blank">&nbsp;</a><a class="refresh" href="#" title="Atualizar lista de Comentários" onclick="jcomments.showPage(6,'com_content',0);return false;">&nbsp;</a></h4>
<div class="comments-list" id="comments-list-0">
		<div class="even" id="comment-item-21">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-21">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(21, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(21, -1);return false;"></a>
<span class="vote-good">+11</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-21" id="comment-21">#</a>
<span class="comment-author">Bruno H. Guimarães</span>
<span class="comment-date">2010-11-03 15:01</span>
<div class="comment-body" id="comment-body-21">Esse artigo está muito 
bom, parabéns. Eu quero saber, porque o fundo de escala do amperímetro é
 ao contrário do fundo de escala do ohmímetro?<br>Se possível me responda pelo e-mail.<br>Grato.</div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(21); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(21,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(21); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
        </div>
	<div class="comments-list" id="comments-list-21">
		<div class="odd" id="comment-item-26">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-26">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(26, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(26, -1);return false;"></a>
<span class="vote-good">+13</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-26" id="comment-26">#</a>
<span class="comment-author">Marcos</span>
<span class="comment-date">2010-12-04 12:41</span>
<div class="comment-body" id="comment-body-26">Porque o fundo de escala do amperímetro é ao contrário do fundo de escala do ohmímetro?<br><br>Ao
 medir a corrente elétrica (amperímetro) num circuito, o valor medido 
inicia-se do zero até o seu máximo. Então o movimento da agulha indica 
do zero até o seu máxímo. Já para medir resistência (ohmímetro) a 
resistência elétrica inicia-se de um valor máximo (infinito OHMs) para o
 zero. Isto é, quanto maior a corrente que passará pela resistência, 
menor é o valor da resistencia, assim temos maior movimento da agulha do
 medidor. Na prática a resistência é o inverso da corrente que percore o
 circuito. Quanto maior a resistência menor será a corrente elétrica que
 percorre o circuito e vice versa.<br>Como o movimento da agulha é 
indicado somente num sentido no medidor analógico, então as escalas 
(ohmímetro e amperímetro) precisam estar desenhadas uma ao contrário da 
outra para que indiquem corretamente o valor a ser medido.</div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(26); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(26,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(26); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
		</div>
		<div class="even" id="comment-item-48">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-48">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(48, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(48, -1);return false;"></a>
<span class="vote-good">+5</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-48" id="comment-48">#</a>
<span class="comment-author">andré luiz</span>
<span class="comment-date">2011-10-03 11:34</span>
<div class="comment-body" id="comment-body-48">meu amigo, é só lembrar 
da lei de ohm, " a corrente eletrica é diretamente proporcional a tensão
 eletrica e inversamente proporcional a resistência", ou seja a escala 
de ohm é contraria as demais. um abração espero ter ajudado.</div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(48); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(48,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(48); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
		</div>
	</div>
		<div class="odd" id="comment-item-27">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-27">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(27, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(27, -1);return false;"></a>
<span class="vote-good">+13</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-27" id="comment-27">#</a>
<span class="comment-author">ana paula</span>
<span class="comment-date">2010-12-16 13:19</span>
<div class="comment-body" id="comment-body-27">Preciso medir a 
resistência de um relé ( relé de motores de geladeiras) no qual a escala
 é bem baixinha ,não consigo medir com um multímetro, gostaria de ajuda 
para saber qual aparelho serviria para medir o circuito do relé ( 
preciso identificar as capacidades )á medição é baixo de 1 ohms.</div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(27); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(27,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(27); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
        </div>
	<div class="comments-list" id="comments-list-27">
		<div class="even" id="comment-item-28">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-28">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(28, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(28, -1);return false;"></a>
<span class="vote-good">+6</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-28" id="comment-28">#</a>
<span class="comment-author">Marcos</span>
<span class="comment-date">2010-12-17 20:50</span>
<div class="comment-body" id="comment-body-28">Ana Paula. Com o 
multímetro (chaveado para a escala de Ohm) você poderá medir o valor 
Ohmico da bobina do relê, como você comentou apresenta ter o valor perto
 de 1 Ohm. As demais capacidades do relê, como a sua tensão de 
funcionamento e a capacidade de condução de corrente nos contatos, se 
não tiver impresso no corpo do relê, você deverá consultar o manual do 
fabricante do relê. Com o multímetro não é possivel medir estas 
capacidades.</div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(28); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(28,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(28); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
		</div>
	</div>
		<div class="odd" id="comment-item-47">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-47">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(47, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(47, -1);return false;"></a>
<span class="vote-good">+1</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-47" id="comment-47">#</a>
<span class="comment-author">andre luis</span>
<span class="comment-date">2011-09-29 13:25</span>
<div class="comment-body" id="comment-body-47">queria saber porque é 
necessário inverter as pontas de prova do multímetro na função de 
ohmímetro para identificar o anodo e o catodo do diodo ?</div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(47); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(47,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(47); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
		</div>
		<div class="even" id="comment-item-49">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-49">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(49, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(49, -1);return false;"></a>
<span class="vote-good">+1</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-49" id="comment-49">#</a>
<span class="comment-author">Antonio Claudio</span>
<span class="comment-date">2011-10-22 17:26</span>
<div class="comment-body" id="comment-body-49">Tenho um multimetro 
analogico (YX-360TRn-a) mas não consigo faser o ajuste para medidas de 
resistências o potenciômetro chega ate o fim e nada<br>ja troquei a 
pilha e a bateria mais não deu certo outro problema é a escala de x10 
segurando nas pontas o ponteiro movimenta indicando que esta escala esta
 queimada como conserta.</div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(49); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(49,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(49); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
        </div>
	<div class="comments-list" id="comments-list-49">
		<div class="odd" id="comment-item-50">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-50">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(50, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(50, -1);return false;"></a>
<span class="vote-good">+3</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-50" id="comment-50">#</a>
<span class="comment-author">Olair Marcos</span>
<span class="comment-date">2011-10-25 09:48</span>
<div class="comment-body" id="comment-body-50">Antônio. É bem provável 
que o multímetro apresenta algum resistor (componente) queimado, abra o 
aparelho e você conseguirá ver o(s) componente(s) chamuscado(s), basta 
troca-los por outros de igual valor. Se estiverem torrados terá que 
recorrer ao diagrama eletrônico que está junto no manual de instruções 
do aparelho para descobrir o valor de cada componente danificado.</div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(50); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(50,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(50); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
		</div>
	</div>
		<div class="even" id="comment-item-52">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-52">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(52, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(52, -1);return false;"></a>
<span class="vote-none">0</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-52" id="comment-52">#</a>
<span class="comment-author">RODRIGO</span>
<span class="comment-date">2011-11-08 16:54</span>
<div class="comment-body" id="comment-body-52">Queimaram dois motores na
 empresa onde trabalho. comprei um multimetro para medir a voltagem que 
esta passando nas tomadas e saber se está além de 220. Como faço esta 
operação?<br>Grato</div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(52); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(52,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(52); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
        </div>
	<div class="comments-list" id="comments-list-52">
		<div class="odd" id="comment-item-54">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-54">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(54, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(54, -1);return false;"></a>
<span class="vote-good">+4</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-54" id="comment-54">#</a>
<span class="comment-author">Olair Marcos</span>
<span class="comment-date">2011-12-07 20:06</span>
<div class="comment-body" id="comment-body-54">Olá Rodrigo<br>Você não 
comentou se o multímetro comprado é digital ou analógico. Suponho que 
você tem um multímetro digital comum, então você quer medir a tensão da 
rede que chega até as tomadas do local. O processo é simples, verifique 
na chave rotativa do multímetro a seção onde temos VAC (Volts para 
corrente alternada) e selecione uma posição acima de 220VAC, geralmente 
os multímetros comuns digitais possuem a posição 750VAC, pode chavear o 
multímetro nesta posição e introduza as pontas de prova na tomada a ser 
medida. O valor lido no multímetro é a tensão AC encontrada na tomada. 
Caso o seu multímetro seja diferente do mencionado, por favor informe o 
modelo para que possamos auxilia-lo da melhor forma.</div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(54); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(54,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(54); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
		</div>
	</div>
		<div class="even" id="comment-item-55">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-55">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(55, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(55, -1);return false;"></a>
<span class="vote-good">+3</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-55" id="comment-55">#</a>
<span class="comment-author">Aridinei</span>
<span class="comment-date">2011-12-10 13:01</span>
<div class="comment-body" id="comment-body-55">Ola amigo parabens pelas 
suas explicaçoes, minha duvida e. eu tenho um multimetro analogico 
gostaria de saber como faço a medida de cargas de pilhas baterias e tal.
 abraços</div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(55); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(55,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(55); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
        </div>
	<div class="comments-list" id="comments-list-55">
		<div class="odd" id="comment-item-62">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-62">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(62, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(62, -1);return false;"></a>
<span class="vote-good">+2</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-62" id="comment-62">#</a>
<span class="comment-author">Olair Marcos</span>
<span class="comment-date">2012-02-05 23:22</span>
<div class="comment-body" id="comment-body-62">A maioria dos multímetros
 analógicos e digitais já possuem uma função para teste de pilhas e 
baterias 9V. Basta selecionar a função apropriada e observar a escala. 
Mas para aqueles que não possuem a função para teste de pilhas e bateria
 o mesmo poderá ser feito medindo a tensão nos terminais, para obter uma
 noção mais precisa das condições da pilha ligue em paralelo com a pilha
 um resistor de 10ohm/3W que servirá de carga. Se a tensão cair abaixo 
de 1.2V a pilha está comprometida. Para baterias de 9V utilize um 
resistor de 27 ou 33ohm/3W. Se a tensão cair abaixo de 8V, a bateria 
está comprometida, sendo necessária a sua troca.</div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(62); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(62,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(62); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
		</div>
	</div>
		<div class="even" id="comment-item-57">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-57">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(57, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(57, -1);return false;"></a>
<span class="vote-good">+2</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-57" id="comment-57">#</a>
<span class="comment-author">toneco</span>
<span class="comment-date">2011-12-28 09:03</span>
<div class="comment-body" id="comment-body-57">Olá, pessoal! adorei o 
conteúdo e a disposição do site. Foi de relevante auxilio ocasionando 
ganho que jamais alcancei pelos sites correlacionados tampouco pelos 
tecnicos que conheço. Gostaria, então, de merecer um favor de qual seja o
 interessado: tenho um multimietro FT-360TR. Ora, na escala DCmA, há 
recurso entre 50 micro-ampères a 250 mA ou 0.25A. De modo que desejo 
testar uma fonte e saber nela qual sua corrente mesmo que nela conste 5v
 e 2A. Pergunta: corro risco de estragar meu multimetro testando uma 
fonte de 2A sendo que sua capacidade (do multimetro) varia entre 50uA a 
0.25A?<br>Obrigado e Feliz Ano Novo a todos!<br>Por favor, me respondam também, se possível, pelo e-mail <span onclick="jcomments.jump2email(57, '449da55f36b545031d166e5fb7d44eea');" class="email" onmouseover="this.className='emailactive';" onmouseout="this.className='email';">toneco.coluna<img src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/email.png" border="0">hotmail.com</span></div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(57); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(57,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(57); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
        </div>
	<div class="comments-list" id="comments-list-57">
		<div class="odd" id="comment-item-61">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-61">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(61, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(61, -1);return false;"></a>
<span class="vote-good">+2</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-61" id="comment-61">#</a>
<span class="comment-author">Olair Marcos</span>
<span class="comment-date">2012-02-05 23:02</span>
<div class="comment-body" id="comment-body-61">Olá Toneco,<br>Na prática
 você não pode medir a corrente da fonte sem ter uma carga conectado a 
mesma, a corrente é proporcional ao consumo da carga conectado na fonte.
 Consta que a fonte é capaz de fornecer uma corrente de até 2A sob uma 
tensão de 5V. Para você poder medir correntes superiores a capacidade do
 seu multímetro com segurança utilize a seguinte técnica: Ligue em série
 no positivo da fonte um resistor de 1 ohm x 10W entre a fonte e a carga
 (quanto mais preciso for o resistor, mais precisa será a medição). O 
resistor ficará ligado somente no positivo entre a fonte e a carga, 
ligue o circuito, agora com o seu multímetro na posição V DC (10 Volts 
DC) conecte a ponta vermelha no resistor, (no lado positivo da fonte) e a
 ponta preta no outro terminal do resistor, agora meça a queda de tensão
 ocasionada no resistor. Exemplo: Se a carga possuir uma resistência de 5
 ohm, a queda de tensão sob o resistor é de 1V então neste caso está 
passando pelo circuito uma corrente de 1 A no circuito. Observe a Lei de
 OHM. Esta é uma forma segura de se medir altas corrente em circuitos DC
 com multímetros digitais ou analógicos de baixa capacidade.</div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(61); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(61,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(61); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
		</div>
	</div>
		<div class="even" id="comment-item-63">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-63">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(63, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(63, -1);return false;"></a>
<span class="vote-good">+1</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-63" id="comment-63">#</a>
<span class="comment-author">Israel</span>
<span class="comment-date">2012-02-09 20:53</span>
<div class="comment-body" id="comment-body-63">Olá, adorei o site, eu 
tenho um multimetro analógico YX-360TRN, eu regulei o parafuso para 
ajustar o ponteiro com o multímetro deitado, mas quando eu coloco ele em
 pé ele sai da posição que eu regulei, e vice-versa, ele está com 
defeito?</div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(63); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(63,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(63); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
        </div>
	<div class="comments-list" id="comments-list-63">
		<div class="odd" id="comment-item-64">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-64">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(64, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(64, -1);return false;"></a>
<span class="vote-good">+2</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-64" id="comment-64">#</a>
<span class="comment-author">Olair Marcos</span>
<span class="comment-date">2012-02-14 21:45</span>
<div class="comment-body" id="comment-body-64">Olá Israel,<br>O 
multímetro analógico é construído para ser utilizado na horizontal 
(deitado) ou levemente inclinado. Utilizando-o na posição vertical (em 
pé) as medições apresentadas poderão ser incorretas, neste caso o ajuste
 que você está tentando fazer ficará sempre fora do ideal. Bom trabalho.</div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(64); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(64,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(64); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
		</div>
	</div>
		<div class="even" id="comment-item-69">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-69">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(69, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(69, -1);return false;"></a>
<span class="vote-none">0</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-69" id="comment-69">#</a>
<span class="comment-author">Edson Rosa</span>
<span class="comment-date">2012-03-18 19:06</span>
<div class="comment-body" id="comment-body-69">Gostei das explicações,sug<span style="font-size:0px;padding:0;margin:0;"> </span>eria que você mostrasse, como se caulcula uma medida no mesmo,se for valores como por exemplo;4,06v. Obrigado e um abraço!</div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(69); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(69,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(69); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
        </div>
	<div class="comments-list" id="comments-list-69">
		<div class="odd" id="comment-item-72">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-72">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(72, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(72, -1);return false;"></a>
<span class="vote-good">+1</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-72" id="comment-72">#</a>
<span class="comment-author">Olair Marcos</span>
<span class="comment-date">2012-03-23 23:57</span>
<div class="comment-body" id="comment-body-72">Caro Edson, Antes de 
tentar iniciar a medição você deverá ter uma noção que tipo de corrente 
AC ou DC vai estar presente no circuito. Entendemos que você pretende 
medir 4,06V num circuito DC, neste caso o primeiro passo a fazer é 
chavear o multímetro para medir tensão DC (DCV), sempre selecione na 
chave de funções uma tensão superior a aquela a ser medida. No nosso 
modelo será a posição 10 DCV. Na escala a observar a leitura será 0-10 
DCV, para ler 4.06V o ponteiro indicará uma posição logo acima do valor 4
 na escala. Observe que para medir tensões DC as pontas devem estar 
corretamente conectadas no negativo para a ponta preta (-) e no ponto a 
ser medido (4,06V) a ponta vermelha (+) do multímetro. Em caso de não 
saber antecipadamente o valor da tensão a ser medido, inicie a medição 
posicionando a chave de funções na tensão mais alta (1000 DCV)para 
depois ir selecionando as opções mais inferiores até o ponto onde o 
ponteiro indique uma leitura aproximadamente no centro da escala.</div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(72); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(72,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(72); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
		</div>
	</div>
		<div class="even" id="comment-item-74">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-74">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(74, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(74, -1);return false;"></a>
<span class="vote-good">+1</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-74" id="comment-74">#</a>
<span class="comment-author">isaias</span>
<span class="comment-date">2012-04-03 15:03</span>
<div class="comment-body" id="comment-body-74">Porque os equipamentos de leitura analógicos normalmente começam com um valor acima de zero ?</div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(74); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(74,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(74); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
        </div>
	<div class="comments-list" id="comments-list-74">
		<div class="odd" id="comment-item-77">
<div class="rbox"><div class="rbox_tr"><div class="rbox_tl"><div class="rbox_t">&nbsp;</div></div></div><div class="rbox_m">
<div class="comment-box">
<span class="comments-vote">
	<span id="comment-vote-holder-77">
<a href="#" class="vote-good" title="Bom comentário!" onclick="jcomments.voteComment(77, 1);return false;"></a><a href="#" class="vote-poor" title="Fraco comentário!" onclick="jcomments.voteComment(77, -1);return false;"></a>
<span class="vote-good">+2</span>
	</span>
</span>
<a class="comment-anchor" href="http://www.eletronicadigital.com/site/instrumentos-medicao/6-multimetro-analogico.html#comment-77" id="comment-77">#</a>
<span class="comment-author">Olair Marcos</span>
<span class="comment-date">2012-04-24 11:35</span>
<div class="comment-body" id="comment-body-77">Caro Isaias, Equipamentos
 analógicos fabricados com componentes de qualidade, fornecem leituras 
com boa precisão também, muitas vezes até superiores aos digitais.<br>Caso
 o seu multímetro apresente uma fuga na placa, defeito no mecanismo do 
ponteiro, ou descalibrado (veja o texto)então poderá apresentar leituras
 erradas.</div>
<span class="comments-buttons">
<a href="#" onclick="jcomments.showReply(77); return false;">Responder</a>
 | <a href="#" onclick="jcomments.showReply(77,1); return false;">Responder com citação</a> | 
<a href="#" onclick="jcomments.quoteComment(77); return false;">Citação</a>
</span>
</div><div class="clear"></div>
</div><div class="rbox_br"><div class="rbox_bl"><div class="rbox_b">&nbsp;</div></div></div></div>
		</div>
	</div>
</div>
<div id="comments-list-footer"><a class="refresh" href="#" title="Atualizar lista de Comentários" onclick="jcomments.showPage(6,'com_content',0);return false;">Atualizar lista de Comentários</a><br><a class="rss" href="http://www.eletronicadigital.com/site/component/jcomments/?task=rss&amp;object_id=6&amp;object_group=com_content&amp;tmpl=component" target="_blank">Sina RSS para comentários ao tópico.</a></div>
</div>
<div id="comments-form-link">
<a id="addcomments" class="showform" href="#addcomments" onclick="jcomments.showForm(6,'com_content', 'comments-form-link'); return false;">Comentar</a>
</div>
<div id="comments-footer" align="center"><a href="http://www.joomlatune.com/" title="JComments" target="_blank">JComments</a></div>
<script type="text/javascript">
jcomments.setAntiCache(1,0,1);
</script> 
</div>
												<div class="moduletable">
					
<!-- /mod_html version 1.0.0.Alpha1-J1.5 (c) www.fijiwebdesign.com -->
 <script type="text/javascript"><!--
google_ad_client = "pub-0131178064072026";
//336x280, bloco-fim-conteudo, 27/01/08, eletdigital
google_ad_slot = "1522063613";
google_ad_width = 336;
google_ad_height = 280;
//--></script>
<script type="text/javascript" src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/show_ads.js">
</script><ins id="aswift_1_expand" style="display:inline-table;border:none;height:280px;margin:0;padding:0;position:relative;visibility:visible;width:336px;background-color:transparent"><ins id="aswift_1_anchor" style="display:block;border:none;height:280px;margin:0;padding:0;position:relative;visibility:visible;width:336px;background-color:transparent"><iframe marginwidth="0" marginheight="0" vspace="0" hspace="0" allowtransparency="true" scrolling="no" allowfullscreen="true" onload="var i=this.id,s=window.google_iframe_oncopy,H=s&amp;&amp;s.handlers,h=H&amp;&amp;H[i],w=this.contentWindow,d;try{d=w.document}catch(e){}if(h&amp;&amp;d&amp;&amp;(!d.body||!d.body.firstChild)){if(h.call){setTimeout(h,0)}else if(h.match){try{h=s.upd(h,i)}catch(e){}w.location.replace(h)}}" id="aswift_1" name="aswift_1" style="left:0;position:absolute;top:0;" frameborder="0" height="280" width="336"></iframe></ins></ins>

<!-- mod_html version 1.0.0.Alpha1-J1.5/ -->
		</div>
	
									</td>
																			<td class="greyline">&nbsp;</td>
										<td width="170">
													<div class="moduletable">
					<h3>Menu principal</h3>
					
<table border="0" cellpadding="0" cellspacing="0" width="100%">
<tbody><tr><td><a href="http://www.eletronicadigital.com/site/pg.html" class="mainlevel">Início</a></td></tr>
<tr><td><a href="http://www.eletronicadigital.com/site/curso-eletronica-digital.html" class="mainlevel">Curso de eletrônica digital</a></td></tr>
<tr><td><a href="http://www.eletronicadigital.com/site/instrumentos-medicao.html" class="mainlevel" id="active_menu">Instrumentos de medição</a></td></tr>
<tr><td><a href="http://www.eletronicadigital.com/site/projetos-eletronica-digital.html" class="mainlevel">Projetos</a></td></tr>
<tr><td><a href="http://www.eletronicadigital.com/site/dicas-tecnicas.html" class="mainlevel">Dicas técnicas</a></td></tr>
<tr><td><a href="http://www.eletronicadigital.com/site/montagem-circuito.html" class="mainlevel">Montagens de circuitos</a></td></tr>
<tr><td><a href="http://eletronicadigital.com/mail" target="_blank" class="mainlevel">Webmail</a></td></tr>
</tbody></table>		</div>
			<div class="moduletable">
					
<table border="0" cellpadding="0" cellspacing="0" width="100%">
<tbody><tr><td><a href="http://www.eletronicadigital.com/site/sobre-nos.html" class="mainlevel">Sobre nós</a></td></tr>
<tr><td><a href="http://www.eletronicadigital.com/site/termos-uso.html" class="mainlevel">Termos de uso</a></td></tr>
<tr><td><a href="http://www.eletronicadigital.com/site/politica-privacidade.html" class="mainlevel">Política de privacidade</a></td></tr>
<tr><td><a href="http://www.eletronicadigital.com/site/contato.html" class="mainlevel">Contato</a></td></tr>
</tbody></table>		</div>
			<div class="moduletable">
					<h3>Feeds</h3>
					<a href="http://www.eletronicadigital.com/site/component/jcomments/?task=rss&amp;object_id=6&amp;object_group=com_content&amp;tmpl=component">
	<img src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/livemarks.png" alt="feed-image"> <span>Feed Edigital</span></a>		</div>
	
										</td>
																	</tr>
							</tbody></table>

						</div>
						<div class="clr"></div>
					</div>
					<div class="clr"></div>
				</div>

				<div id="whitebox_b">
					<div id="whitebox_bl">
						<div id="whitebox_br"></div>
					</div>
				</div>
			</div>

			<div id="footerspacer"></div>
		</div>

		<div id="footer">
			<div id="footer_l">
				<div id="footer_r">
					<p id="syndicate">
						
					</p>
					<p id="power_by">
	 				 	Produzido em <a href="http://www.joomla.org/">Joomla!</a>&nbsp;-&nbsp;
						<a href="http://www.eletronicabasica.com/">Eletrônica Digital</a>&nbsp;© 2010.
					</p>
				</div>
			</div>
		</div>
	</div>
</div>




<div id="fbox_world"><div id="fbox_bg" style="opacity: 0; position: fixed; left: 0px; top: 0px; width: 1267px; height: 705px; display: none;"></div><div id="fbox_progress" style="position: fixed; left: 621px; top: 340px; display: none;"><img src="Mult%C3%ADmetro%20anal%C3%B3gico-pag-1-de-3_arquivos/indicator.html"></div><div id="fbox_fg" style="width: 402px; left: 426px; top: 40px; height: 612px; position: fixed; display: none; opacity: 0;"><div id="fbox_fg_inner"><div style="visibility: visible; width: 396px; overflow: hidden; height: 580px;" id="fbox_content"></div><div style="visibility: visible; width: 402px;" id="fbox_spacer"></div><table style="visibility: visible; width: 402px; height: 17px;" id="fbox_bar" border="0" cellpadding="0" cellspacing="0"><tbody><tr><td><div id="fbox_sys"><nobr><a style="visibility: visible;" id="fbox_resize_a" href="#">Maximize</a><span style="visibility: visible;" id="fbox_resize_nbsp">&nbsp;&nbsp;&nbsp;&nbsp;</span><a id="fbox_close_a" href="#">Close Here</a></nobr></div><div style="float: left; visibility: visible; height: auto; overflow: visible; width: 155px;" id="fbox_title">Multímetro analógico Sanwa</div><div style="margin-top: 3px; visibility: hidden; display: none;" id="fbox_navig"><nobr><a id="fbox_prev_a" href="#">« Previous</a>&nbsp;&nbsp;&nbsp;&nbsp;<a id="fbox_next_a" href="#">Next »</a></nobr></div></td></tr></tbody></table></div></div></div></body></html>